Why Is Your Tuttnauer Autoclave Displaying A Low Temp Error Code?
When your Tuttnauer autoclave begins flashing error codes your first response is almost certainly panic. Your autoclave is among the maximum vital items of apparatus you have got and with out it you’ll’t sterilize your gear. And if you’ll’t sterilize your gear you might be into bankruptcy. Don’t fear regardless that. A Low Temp error code is not any motive for alarm. And you’ll almost certainly restore the issue your self.
A Low Temp error code is displayed if the temperature within the chamber begins to drop. If the temperature drops greater than 4.5ºF the sensor will mechanically abort the cycle and the Cycle Fail indicator lighting may also flash. All the ones flashing lighting make the placement look like the aircraft is set to crash and everybody must grasp a parachute. But it is just a bit turbulence. Hang on. You can repair it.
First, test the water stage within the chamber. If there may be no longer sufficient water within the chamber then there would possibly not be sufficient steam and steam is essential to create the warmth that sterilizes your gear. Here are a few things to test:
- Check the water stage within the reservoir to verify it is complete
- Be certain your autoclave is correctly leveled
- Check the water sensing electrode for filth or it can be shorted out
- Make certain the water pump is not clogged
- Look for clogged water traces
- Make certain the air outlet valve is operating correctly
Next, if the water stage within the chamber is okay, test your cycle time. If the cycle used to be set for too lengthy a time period the water within the chamber can have boiled out and that’s the reason why the chamber is dry. This occurs regularly, particularly on autoclaves with handbook set timers.
It it is not a timer downside, then you definitely want to test the security thermostat. It is also faulty. When the security thermostat opens it mechanically turns of the heating parts – as a security precaution. In the Tuttnauer T935N5 and T935N6 autoclaves the security thermostat has been recognized to open upfront.
You must by no means try to recalibrate the security thermostat. It used to be calibrated on the manufacturing unit and must get replaced with every other manufacturing unit calibrated thermostat. That stated, you’ll substitute that thermostat your self.
Unplug your autoclave first after which take away the wires from the terminals at the thermostat. You’ll need to tag them so you’ll simply reconnect the alternative. Next, take away the screws or unscrew the lock retaining the security thermostat to the mounting bracket. Loosen the bolts on that hang the thermostat to the heating component and take away the probe. Be cautious when uncoiling the tubing that connects the probe to the touch frame. It’s full of liquid and in the event you puncture it you’ll be able to want to substitute it, too.
Now, simply mount the brand new thermostat and run a check cycle to verify the whole thing’s running correctly once more.
As you’ll see, there may be in point of fact no want to panic when your Tuttnauer autoclave presentations a Low Temp error code. It’s simple to fix the issue your self. In truth, it is simply as simple to maintain almost about any downside you have got along with your autoclave. So do not let all the ones flashing lighting scare you. Pick up a wrench and fasten it your self.